Ohlone doesn't have a bio yet.
Humidity Sensor fixed
By: Ohlone , 6:11 PM GMT on February 10, 2017
Well I'm on my third Temperature Humidity sensor. Seems all the rain soaked the sensor and broke it. New one came in three days from Davis Instruments.
I've inverted the sensor mount for the Davis VP2 with fan aspirator. It looks like its back and working again. Now we just need for WU to work with the folks at Ambient VWS (Ed) to get the support for the mesonet data and forecasts working again. Realize WU just moved their servers.
We hit a record rainfall this year since I have been running my station. Over 71" rain.
The views of the author are his/her own and do not necessarily represent the position of The Weather Company or its parent, IBM.
No reader comments have been posted for this blog entry yet.